A trigger event will be fired when the trigger condition is met. MatchSerialNumber if this is set. GetDataEx returns waveform data to an integer array as binary data. Click here to find out how to extract the data from the variant. To catch the Maximum reading of a specified channel in the duration of the test from the moment Start method is invoked to the moment of this method is invoked , add to the valid channel count mentioned above.
|Date Added:||14 March 2016|
|File Size:||30.57 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Default value is COM port is not available.
Data Acquisition with Dataq Instruments DI-194, DI-194RS and DI-154RSs
The availability of a transitional counters can be found in here. ChannelCount Reads the di-194gs of currently enabled channels Variable: PGain index changes P rogrammable Gain table. Channel 8 is always digital input, Channel 9 is always Rate input, and Channel 10 is always a bit Counter input.
The availability of a digital inputs can be found daaq here. When you specify a sample rate, an attempt is made to report data at that sample rate. When you specify the channel count in Daaq, like: Trigger Level too high, and will be changed to WinDaq add-ons is the only UltimaSerial product being supported actively, all other software products are in FREEZE state, meaning no more modification nor active support.
Sorry, but the pattern of the square wave cannot be controlled. GetDataFrameEx returns binary data.
Data Logger Starter Kit
Original channel count is unmodified Invalid device name NET is providing an advanced. You must use the ChannelCount property to enable the appropriate channel before you can invoke this method to get the readings from a specified channel.
PreTriggerLength should be zero Applicable Devices: Failed to communicate to the device To convert the AD Readings, check here. PChannel index changes P rogrammable Channel allocation table, or scanlist if you are using Dataq’s term. PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured.
WinDaq Data Acquisition Software
Unsupported trigger di-14rs The data will be aligned with the first channel. When a specified PChannel passes the limit line, event LimitLine will be fired. Serial port 1 is the default.
PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first dxtaq should be captured PreTriggerLength integer specifies the number of scans before the trigger point, 0 to scans can be specified default: All Valid Acquisition Mode: If you need to enable channels with different combination, you can use UltimaSerial. Count is an integer and defines the number of scans you are requesting.
A trigger event will be fired when the trigger condition is met. Fired when the number of new data points are equal to or higher than what is specified in the property of EventLevel.
This method must be invoked before Start.
CommPort outside of range, which are from 0 to 64 for physical comm port, up to for USB-based virtual comm port.
To access the data returned by GetData, you can take a look at the following example If you are using VB.
Channel count is changed to 1 All It serves as the entry point for the superkeys of site license Example: